Connect the instrument or to the electric network (220 W) or to a battery (12 W).
Simultaneous power supply from the network and the battery is not recommended.
Connect the red clamp of the cable to a plus battery terminal, the black one connect to minus battery terminal
Connect the instrument cable to the instrument
Switch on the instrument
The instrument identifies the power source
To start measurements, press «Start/Stop»
Determination of current in the winding
Measurement is in progress
Replacement of three arrows ↓↓↓ by two ↓↓, then by one ↓ means continuous resistance reduction. Hence, the process of resistance identification is normal.
If arrows are replaced by a rhomb ♦, the measurement can be stopped by pressing «Stop»
The measurement is over. The readings are shown.
For repeated measurement press «Start/Stop»
To go back, press «Cancel»
Disconnect the cable
Switch off the instrument
Disconnect the instrument from the network
Disconnect the instrument from battery